All operating data have been factory set in accordance with your specifications. The selected creepage distances and clearances have been sized in conformity with VDE and IEC for contamination category 2. These concise instructions do not include the following: Page 8 Connection of shields to the IFC 1 DS Signal cable 2 Shield terminal for shielded signal cable 3 Field power cable 4 Shield terminal for shielded field power cable 5 Housing wall, signal converter 6 Cable entries 2 Electrical connection of outputs 2. Page 15 Function 3. Page 11 Standard factory settings Function 1. Supply circuits and output circuits are designed to meet the standards krobne overvoltage classes III and II, respectively.
|Published (Last):||9 September 2008|
|PDF File Size:||9.54 Mb|
|ePub File Size:||14.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
The primary head must be installed in the pipeline as described in the installation instructions inside the packing of the primary head. Page 4: Versions Ifc Signal Converter Display version option with local display and control elements. All operating data factory-set to your order specifications. If flowmeters need to be returned to Krohne, please note the information given on the last-but-one page of these Instructions.
Krohne regrets that it cannot repair or check your flowmeter s unless accompanied by the completed form sheet. Therefore, when installing, ensure primary constant GKL is identical; The selected creepage distances and clearances have been dimensioned in conformity with VDE and IEC for contamination category 2. Page 9 Stripping preparation of signal cable A 1. Page Cable Length max.
Please also refer to Sect. Page 13 Connection diagrams for outputs 2. All operating data have been factory set in accordance with your specifications.
Please refer to Sect. Page Factory Settings Factory settings All operating data are factory set according to your order specifications. If you have not made any particular specifications at the time of ordering, the instruments will be delivered with the standard parameters and functions listed in the Table below.
KROHNE IFC 010 D PDF
Page 3 1 Electrical connection: If installed and operated in accordance with these operating instructions, it will rarely present any problems. Signal cable to be permanently laid. Vo r tex flowmeters. The operator is, howeverobligated to take note of these sections in the. Switches, counters, displa ys and recorders. P Pulse output overranged. Setting ranges The ranges are dependent on the meter size DN and the flow velocity v: These concise instructions do not include the following: Lev el measuring instruments.
KROHNE IFC 010 Converter FR Manuel d'utilisation